Bloomberg News, Bloomberg Television, California Superior Court, Cobalt chrome, Cobalt poisoning, DePuy, DePuy ASR Hip, depuy hip recall, DePuy Hip Recall Litigation, DePuy Lawsuit, DePuy Orthopaedics, FDA, free legal consultations, Hip Replacement, hip replacement failure, hip revision, James Anderson, Johnson, Johnson & Johnson, joint replacement, litigation counsel, Los Angeles, metal, Metal on metal hip, metal-on-metal hips, metallic debris, metallosis, Orthopedic surgery
DePuy ASR Trial Update: J&J Engineer-Suggested Redesign Meant to Minimize Metallic Debris Never Implemented, Parker Waichman LLP Reports – SFGate.
This week, a bioengineer for Johnson & Johnson’s DePuy Orthopaedics unit testified in Los Angeles state court in the first of 10,000 DePuy ASR lawsuits to go to trial. According to Bloomberg News, James Anderson worked on redesigning the ASR for three years before the project was ended; even though he found ways to potentially lower the amount of metallic debris released by the metal-on-metal hip implant, no changes were ever implemented.
(PRWEB) February 08, 2013
Parker Waichman LLP, a national law firm dedicated to protecting the rights of victims injured by defective medical devices, recounts the latest DePuy ASR trial developments reported by Bloomberg News. Bloomberg News reported on February 7th that James Anderson, a bioengineer for Johnson & Johnson’s DePuy unit, testified in Los Angeles state court at the trial of Loren Kransky, whose DePuy ASR lawsuit is the first of 10,000 to go to trial (Kransky v. DePuy, BC456086, California Superior Court, Los Angeles County). According to his videotaped testimony, Anderson worked on redesigning the ASR hip implant for three years before the project was ended. Anderson testified that he found ways to decrease wear on the cup, but no changes were ever made to the now-recalled metal-on-metal hip implant.
“Again, we have evidence that Johnson & Johnson was aware of problems with the ASR hip implant prior to the recall in 2010,” said Daniel C. Burke, Senior Litigation Counsel at Parker Waichman LLP. “Despite knowing that this hip replacement has a high rate of wear, J&J did not actually make the necessary changes to try and improve the ASR.” Parker Waichman continues to offer free legal consultations to hip replacement patients implanted with the DePuy ASR, along with other all-metal hip replacements. The firm advises hip replacement patients to speak with their surgeons if they experience symptoms such as:
- Difficulty walking, or a change in your ability to walk
- Popping, creaking, or other sounds emanating from the area of the implant
- Metallosis, high levels of metal ions
- Early failure, revision surgery
According to Bloomberg News, this week jurors heard a videotaped testimony from Anderson stating that he was disappointed when the project to redesign the DePuy, known as “Project Alpha,” ended in 2008 without any changes to the all-metal hip implant. Anderson joined a team to study the “next generation ASR” in 2005. On April 7, 2008 he wrote to a superior, stating that “a small improvement to geometry could represent a large improvement for many patients.” In a separate document he wrote that by removing the groove on the ASR cup, wear on the cup “is approximately threefold less than the original design with the groove.”
Johnson & Johnson never implemented any changes to the ASR hip implant, which was recalled in 2010 due to a high failure rate. At the time of the recall, the company said that the implant failed in 12 percent of patients within five years. More recent data from an Australian national joint registry showed that the failure rate exceeded 40 percent.
This week, jurors also heard an expert witness testimony from a McGill University professor who stated that the ASR implant had a number of design defects causing it to fail more than other hip replacements. He said that cup shape, which is less than half a circle and has thin walls, attributed to the defective nature of the implant.
The U.S. Food and Drug Administration (FDA) has issued new guidelines for patients implanted with metal-on-metal hip implants such as the DePuy ASR. In a Safety Communication issued January 17th, the agency advised physical examinations, diagnostic imaging and metal ion testing as needed. The FDA is also looking to tighten regulations on all-metal hip replacements that would require manufacturers to prove the devices are safe and effective before selling them.
Parker Waichman LLP continues to offer free legal consultations to victims of metal-on-metal hip implant injuries. If you or a loved one experienced premature failure of your implant or other health problems associated with a metal-on-metal hip implant, please contact their office by visiting the firm’s Defective Hip Implants page at yourlawyer.com. Free case evaluations are also available by calling 1 800 LAW INFO (1-800-529-4636).
Parker Waichman LLP
Gary Falkowitz, Managing Attorney
Read more: http://www.sfgate.com/business/prweb/article/DePuy-ASR-Trial-Update-J-J-Engineer-Suggested-4264231.php#ixzz2KRdsVdyA
- DePuy ASR Trial Update: J&J Engineer-Suggested Redesign Meant to Minimize Metallic Debris Never Implemented, Parker Waichman LLP Reports (prweb.com)
- DePuy ASR Hip Lawsuit Update: Bernstein Liebhard LLP Weighs in On Latest Development in DePuy ASR Recall Trial (prweb.com)
- DePuy ASR Hip Lawsuit News: Expert Testifies on Toxicity of Metal Debris, Notes Bernstein Liebhard LLP (earlsview.com)
- DePuy ASR Hip Lawsuit News: Expert Testifies on Toxicity of Metal Debris, Notes Bernstein Liebhard LLP (prweb.com)
- During Trial, New Details Emerge on DuPuy Hip – NYTimes.com (earlsview.com)
- J&J Failed to Weigh Hip Metal Debris Risk, Witness Says – Bloomberg (earlsview.com)
- J&J Had ‘Inappropriate’ Risk Controls for Hip, Jury Told – Bloomberg (earlsview.com)
- J&J Failed Own Safety Test in Hip Design, Witness Says – Bloomberg (earlsview.com)
- Hip Implant’s Risks Inadequately Assessed, DePuy Report Found in 2010 – NYTimes.com (earlsview.com)
- Maker Aware of 40% Failure in Hip Implant – NYTimes.com (earlsview.com)
Pingback: Tim’s Horror Story – DePuy has more Problems with Ceramic on Metal Pinnacle Hip – and another Surgeon shows no concern … « Earl's View
Pingback: What DePuy Orthopaedics Knew About Its All-Metal Hips – NYTimes.com « Earl's View
Pingback: J&J again recalls thousands of faulty hip implants – US News and World Report « Earl's View
Pingback: “What a Mistake,” says DePuy Hip Recipient « Earl's View
Pingback: J&J Feared Metal Backlash Would Hurt Hip Sales, Jury Told – Bloomberg « Earl's View
Pingback: Patients victims of untested medical devices – SFGate « Earl's View
Pingback: Business: Washington Post Business Page, Business News « Earl's View
Pingback: DePuy Trial Continues Despite Talks of Settlement – Florida Mass Tort Attorneys, Lawyers | Searcy Mass Torts « Earl's View
Pingback: DePuy ASR Hip Recipient Feared for His Life before Revision Surgery | Earl's View
Pingback: DePuy Pinnacle Hip Replacement Lawsuits: Bernstein Liebhard LLP Comments on New Study Finding that Hip Replacement Failure More Likely in Women | Earl's View
Pingback: DePuy ASR Recall Lawsuit Update: Rottenstein Law Group Responds to ASR Hip Implant Designer Who Testified He Would No Longer Implant His Patients with the Device | Earl's View